00:45 UT: alarm through DBBC has changed in last digit. Might cause one of these small cklock jumps. (Lucia)
02:07 UT: dbbc kept drifting. Schedule halted, dbbc reset, fs restarted. First good scan 239-0210. Missed 90 GB. (Lucia)
09:57 UT: ALARM: Delay through DBBC has changed. Monitor for stability and reset DBBC if it drifts. Delay is not stable, dbbc keeps drifting, schedule halted and dbbc reset. Missed scans 239-0957 and 239-1007 (Ross).
Disk VSN: HOB+0083
Data volume at beginning: 0 GB
SETUP: ntpq has highly unstable offset in fifth row ranging between -13 and -22 ms. First two are stable, and third and fourth are consistently less than 10 ms (Ross).
SETUP: Cannot record to HART+101 module in Bank A. Module HOB+0083 in bank B can successfully be recorded to (Ross).
/home/www/auscope/opswiki/data/pages/handover/aov004_and_rd1507.txt · Last modified: 2015/08/27 10:13 by David Horsley