skip to content
AuScope VLBI Operations Wiki
You are here:
Documentation for operators
Fringe checks not performed otherwise started on time.
Frequent error: “Error being ignored: error qk -211 tsys value for device id overflowed or was less than zero.”
Bands 07 - 12 high, approximately 130 - 140K.
Recording to BKG+0133.
power failure caused antenna stuck & missed scans: 238-0301 - 238-0333b, @ 03:00-03:40; clkoff reading was wrong. restart mk5 necessary.
Scans missed/large clkoff: 238-0040b - 238-140b (00:40-01:40 UT). first good scan again: 238-0142 @ 01:42:32
missed scans occurred after schedule power failure. Generator jumped on, but we got an awkwardly high delay difference.
Tsys in band 09 greater consistently than 190K.
Recording to WSRT-049.
eRemote connection failed within 5 minutes of starting and on numerous other occasions for up to half an hour.
Frequent warnings : “2014.237.19:37:27 WARNING: error s1 -200 odd response from the pic. aborting….” and “2014.237.19:37:29 WARNING: error s1 -201 ifpcn: error response from ifpic, see s1 -200 error for text.”
Recording to OSOD-054.
You could leave a comment if you were logged in.
· Last modified: 2014/08/26 03:41 by
Back to top