Disk VSN: USN-0137/4000 | Data volume at beginning: 0.0 GB
1700UT Experiment started okay (Lucas).
clkoff-maserdelay is now -0.986 us at Katherine despite no sync error.
dbbc=pps_delay reported 61881 rather than 61950. Did a dbbc=pps_sync, then fmset. This now fixed both the clkoff-maserdelay and pps_delay. Will introduce clock jump however so consid20er scans before 119-1712 missed (Lucas).
2020 UT Field system crashed, setting off watchdog alarm. Restarted field system and put schedule back on, missed scan 119-2009a to 119-2035b (Lucas)
2040 UT Pretty serious lag between commands and response from field system, not a ping issue (Lucas).
ALARM: scan_check reports mark5 data format problem. Module swap probably required.
Monitoring. Could be a one off.
Didn't recur in following 2 scans.
After reset systemp12 magically worked again too. Annoyingly inconsistent and unexplained (Lucas).
03:05 UT pps delay drifting, dbbc reconfigure, missed scans 120-0306b and 120-0310 (warren)
03:16 UT schedule resumed (w)
07:10 UT Zero values return in response to wth command(Simin).