Disk VSN: USN-0095 | Data volume at beginning: 12 GB
Note: Record using jive5ab, please do not start DIMnino during next experiment setups or it will crash the mk5. If need to use dimino, go to pcfsyg and ctrl+c on the upper right terminal and run pkill jive5ab before starting it.
TPI overflow problem again on pointing check. Restarted dbbc software a few times, dbbc=pps_sync sent a few times, didn't work.
1700UT Experiment started okay (Lim).
1700UT error m5 -900 not whilst recording and error m5 -906 mark5 return code 6: inconsistent or conflicting request seems to happen on every scan.
The above error is due to mk5=bank_set? being called during midob. Commented out mk5=bank_set? in midob and moved it to preob. Temporary fix documented here (Lucas)
/home/www/auscope/opswiki/data/pages/handover/r1913.txt · Last modified: 2019/09/24 12:52 by Tiege McCarthy