skip to content
AuScope VLBI Operations Wiki
You are here:
Documentation for operators
This is an old revision of the document!
recording to HOB+0119 start at 8692
recording to BKG+0126 start 0
. Test record OK.
After start channels 11, 12 looked suspect in autocorrelation.
clock jumped at 19:00. halted schedule and reconfigured dbbc at 19:03. fs crashed also.
one missed scan 240-1902a. next good scan 240-1916. (missed scan record stop means -20
clock delay drifted from -0.29 us to -0.4 us around 240/20:30 and has stabilised see ke_ddiff.txt plot
missed scans 240-2312 and 240-2328 due to dbbc reconfig.
clock delay stable since 240-2330 (as of 240-0315)
recording to USN-0019 start 0
spurious wind stow 240/21:22, antenna will not release elevation brake. Sending Jack from moblas to try handbox antenna alive
also note anemometer cant connect to device
schedule halted 240/21:36
drives still inoperable at 240/23:59
restarted for scan 241-0447
You could leave a comment if you were logged in.
· Last modified: 2014/08/29 04:49 by
Back to top