0102 formatter to fs time difference was 2 secs, fixed using fmset when telescope was not recording, as per wiki guidelines (AK).
0331 UT Katherine threw two errors, consulted with Ellen Manning, errors are listed below (AK)
03:31:20 WARNING: error an -103 pointing computer tracking errors are too large.
03:31:20 WARNING: error qo -301 warning: onsource status is slewing!
0524 UT katherine threw the same two errors listed in the comment above (AK)
mk5=dot? time was slightly high, approx 13 ms. Consulted with Jamie who suggested running “sy=run setcl offset & ” when no data is being recorded. Time was constant around 13 ms even after running the command (AK)
Module: HAY-0028, start pos: 0 GB
1730 UT Experiment started OK (JS)
11:30UT: Yarragadee want to be called when experiment is finished (18:00) to change over modules (NH)
/home/www/auscope/opswiki/data/pages/handover/t2109.txt · Last modified: 2016/02/17 11:41 by Jim Lovell