Sometimes the DBBC needs reprogramming because channels have been dropped or spectra look strange. To do this, preferable in a gap between scans, follow the following steps:
Applications --> Auscope Hobar --> VNC -->dbbchb
DBBC2 Control DDC v104_2.exe. Close it and then restart it from the desktop. When prompted to reconfigure press “y”, it takes about 2 minutes.
You should then see a row of LEDs on the DBBC flash in sync, once a second.
In the interface, type
S followed by
y, then press the ESC key.
and check that you see
FHG_off). If you see something like
fmset may need to be run again.
clkoffcommands which read the Agilent counters.
ERROR sp -4 GPIB Device time-out on response c2
To fix this, try re-setting the counters. In the
oprin window, type
and try the
maserdelay commands again.