In Attendance: Nick, Lucia, Jamie, Jonny, Bryn, Jesse, Ross
Apologies: Tiege, Liz, Lucas, Jim
The observer on from 9-5, please check if the correct modules are loaded in for the night?
Meeting roster is still pretty much empty, fill it!
Issues from the logs
rd1702-ho26m-getting error qy -1 error in response from counter. Tried restarting with 'counter' command. No response when issuing clkoff and maserdelay commands. Called on-call (Warren), to attempt to fix (Bryn) and WARNING: error ib -4 gpib device time-out on response ca. Error likely related to previous, trying to reset with counter does nothing. Manual reset of GPIB device might be needed (had a similar issue to this a few months ago) Waiting until staff on site to fix. (Bryn) - Hardware failure, not critical if you can read the clkoff
r1777-hb12m- During scan 031-0938 the power for the media converter died. Required me to drive out to the telescope and call Brett who talked me through it. (JR) -solutions seems to be something that Brett needs to sort out as per his email.
aua016-yg12m- mk5 had crashed and needed a few power cycles before it came back online. Is there a better procedure for this? -kvm console
/home/www/auscope/opswiki/data/pages/operations/agenda2017_02_02.txt · Last modified: 2017/02/02 02:12 by Arwin Kahlon