yg: Assigned module (HOB+0107) was full, but used HOB+0097 instead. Module allocation spreadsheet wasn't updated in time.
yg: Eremotectl connection dropped out, caused experiment to start late. Network reliability issue, Brett might take a look at it.
If eremotectl dies, can try restarting ercd daemon on pcfs machine.
If network connection completely drops out call Yg, and they can reboot router from their end.
yg: mark5 time 0.2 second off from fs time and computer time, but delays were stable. Experiment already started, didn't want to introduce a clock jump so just left it. Probably due to ntp time small errors, should correct itself.
ke: Unlabelled module (StreamStor1) and too dark to verify. Shouldn't be an issue now that Ke interior light fixed
Ke interior light fixed. Can turn on at pdu1ke.phys.utas.edu.au. If stops working, probably bulb burned out, let Mick know (auscopeke mailing address).
Hobart26m drives major failure. Encoder has failed, in repair. Need to do new pointing once it's fixed. Out of action for at least this week.
slogit script is fixed on all machines
Use comments.py or comments.sh? comments.py is best if it works, but shell script is a backup
/home/www/auscope/opswiki/data/pages/operations/agenda2018_04_05.txt · Last modified: 2018/04/05 03:12 by Patrick Yates