Attended: Tiege, Mas, Atifur, Katie, Lucia, Simin, Jay
Absent: Patrick, Ross, Jesse
Katherine maser/dbbc/heating appears to have been successful. No more clock jumps recorded in past week?
Antenna monitor says YG stowed even though tracking and onsource (r4861, r1862), az el still move just little box says stow
00UT session postponed - probably not until 2020?
R4860 - lots of issues with mk5 and module but they appear resolved and/or specific to that module. Laggy HDD in drive caused mk5cn errors (alarms) despite DIMino running and responding. The timeout lag caused the fs to think mk5cn was not responding (in 5ms period over which it checks?)
AUM010 - HB 'Network to the antenna failed. Fixed by power cycling media adaptor, then rebooting antenna controller'. How does one repeat this fix? Ask Jamie another time
R1861 - KE 'Experiment started late due to bad module (USN-0191/2000). First scan 268-1801b. (Atifur)' - What was the diagnostic for this? How do we see this early and prevent it? (did you do a test recording?) - Tried normal things, dimino check, restart mk5.
CRDS98 - needed to manually download and drudg .vex file as the slogit scripts only recognise .skd files?
/home/www/auscope/opswiki/data/pages/operations/agenda2018_10_04.txt · Last modified: 2018/10/04 03:16 by Lucas Hyland