This is an old revision of the document!
Hobart 26: Formatter to FS time 0.5s or greater. Yr12: Back to regular Mk5. DBBC PCI 7200 card replaced. sync_error fixed.
AOV029: Hb26: Newsmerd went down. Manually changed receivers and tried to boot newsmerd on site without success. Stuck in a weird state. Live page not working, cannot access vdesk or bruce. Lucky was able to stow antenna remotely with field system on hobart after remote boot, but could not confirm successful stow until drive out. Started fs-mk5hb on pcfshb, still is running there. mk5hb required a reboot too. Scan 326-2208 missed after halting schedule - mistook the receiver p0sition as incorrect R1869: Yg12: 03:32 Too many “mk5cn time out” errors restart mark5 recorder 03:47 DIMino crash? restart Dimino
OHG116: Ke12: 0959 UT WARNING: error ib -402 gpib(enol) write handshake error, check instrument and cable, w1