Please remember to create handover note page during setup.
No connection to Yg and Hb maser.
Handover note created without experiment name.
“Experiment started” information not provided.
Data volume at beginning not specified.
Ho: 1307UT - fmset puts the field system time 6.2 seconds ahead of the Mark5 and Computer time, this seems to result in scans ending (and starting) ~5 seconds ahead of schedule (when going off the log time). We are in the last 50 mins of the experiment so I am going to leave as is rather than spend 25 mins restarting machines and end up losing more data than would be (potentially) saved. (Tiege)
Yg: pcfsyg disk filled up 15 minutes before experiment started. Needed to remove some old epi/png files from the auto_corr_test directory. (Tiege)
Ho: 1725UT - Cryo monitoring system not working at Ho (ERROR ma -4 Device rx timed-out on response). I have replaced the cryomon and rxmon procedures with nops in EXPERIMENT procedure library for the time being, because otherwise it waits for each rx measurement to time out. I have also commented out the rxmon command in the midob procedure in STATION library, as that is also waiting for timeout, and screwing up scan timing. (Patrick)
Fixed by resetting TTY distributor with “reset” switch on old S/X MarkIV rack. Serial communications from Hobart to s/x receiver are still patched through this rack. (Warren and Brett)
1725UT - wx command not working. (Patrick)
/home/www/auscope/opswiki/data/attic/operations/agenda2019_05_16.1557973844.txt.gz · Last modified: 2019/05/16 02:30 by Lim Chin Chuan