Attended: Lim, Katie, Warren, Pradyumna, Tiege, Jamie, Mas, Jay, Patrick
Ho gives continuous error ma -4 device rx timed-out not response.
It might be related to receiver package. Occasion ones should be fine. Continuous ones should be considered and restart fs. Via vdesk try to bring back the antenna to the schedule. Sometimes have to try drvon for 5 min! Brett and Jamie will investigate this.
Ho sometimes has a problem with DIMino running. For Ho setup, if there is a problem with DIMino, please check whether Jive5ab or other programs are running or not.
If you need to upload scripts for an experiment, please let Jamie know about them.
Jamie will be on parental leave from next week for one month.
Network issues (Jamie)
The communication was on/off for few hours for this experiment and after that it was ok.
mk5 connection closed. It might be a problem with the recorder or module (Lim)
Via vdesk and drvon will bring back the antenna to the schedule. Sometimes we have to try several times. Drives need to be fixed.
continuous error ma -4 device rx timed-out not response (Mas & Lim)
Config & mark5B issues (Jamie)
It was a rush up to start the experiment that causes these issues.
lost antenna comms. antenna comms restored (Warren)
The “antenna=open” command solved the problem.
Problem with the drive. Warren needs to restart the drive manually (Mas)
continuous error ma -4 device rx timed-out not response. terminate the PCFS and halt schedule. Received error “Allocating memory for antenna monitoring”. Couldn't connect to vdesk even after rebooting the sys26m (Mas)
Brett used 'source=disable' to reestablish communications (Jamie)
Source not reached. Encoder glitch? (Jamie)
/home/www/auscope/opswiki/data/pages/operations/agenda2020_07_16.txt · Last modified: 2020/07/16 04:10 by Simin Salarpour