onsourcecommand to check the antenna is where it should be,
disk_posin eRemoteControl should report two values - the current number of bytes recorded, and the number of bytes at start of previous scan.
rxpShould be -256, +248, +18 (x,y,z). The values may be off by 2-3 mm which is the measurement noise. The 26m live page will have the receiver listed (left side)as “S/X” if all is well.
cryomonwill return information on the S/X cryogenics. The lines to look out for are the
pv = nrtthe pressure will go down on a cold night. It is advisable to only call Brett if the pressure is quite below 275 and or there have been massive sudden dips in pressure indicating a leak. The plot might be up in the vnc newsmerd session (Desktop 4).