flagr/antenna,acquiredor watch the system monitor display). While you're waiting, Check the generator status:
You will see what the current Conditioning module settings are. Output format is:
<time>/<Module label>/<IF input number>,<Auto or Manual gain control>,<Nyquist filter number>,<Target power level>,<attenuation>,<actual power level>
Check there’s agreement with what appears in the
ifd8f definition in the procedure file. The actual power level should agree pretty well with the target level. The attenuation number can be anywhere between 0 (none) and 63 (maximum). If you see it at 0 or 63, it means the Conditioning module is having trouble getting the power to the right level. You may want to adjust the attenuators (see configuring the RF and IF signal paths) to get them back in range.
You will see what the DBBC has set the BBC freqs to (compare with the .prc file). Output looks like this:
<time>/<bbc name>/<Frequency (MHz)>/<Conditioning module in use>,<Bandwidth (MHz)>,...
the Frequency, Conditioning module label and bandwidth should agree with the listing in
and check the difference reported in the log monitor
flagr/antenna,acquiredbefore continuing to the next:
test1 test2 test3 test4 test5 test1
These commands send the antenna to the following set of (Az, El) coordinates in turn : (1, 87), (1, 6), (1, 45), (-269, 45), (269, 45), (1,87). The idea here is to exercise the antenna and log the results se we can look for any changes over the days that may indicate mechanical problems.