Ho Module: JIVE-064, start pos 0 GB 4:44UT - small power glitch caused Mark 4 rack to reset some values, setupsx was run again and fixed the problem (Imogen)
Some how the log was set to station.log at 27.13:19, and most of the experiment was recorded to that log file. Copied out the appropriate for final log.
Hb Module: BKG+0133, start pos 0 GB
Ke Module:USN-0131, start pos 0 GB
DBBC not behaving, so couldn't observe (Ellen)
Yg Module: USN-0175, start pos 0 GB
Jamie tried to fix the power levels but couldn't. As a results, tsys bands 05 -12 are all bad. (Ellen)