Disk VSN: MET-0001/2000 | Data volume at the beginning: 0.0 GB
18:30:00 UT Formatter to fs time offset was high. Rebooting mk5 and sync did not fix the issue. Had to wait for Warren to fix it. Scan missed from 151-1830b to 151-1859. Recording resumed from 151-1903. (Atifur)
19:35:36 UT No delay drift was noted but alarm sounded with error “Delay through DBBC has changed”. Schedule halted and DBBC reconfigured. Schedule resumed at 19:42:16. scan missed: 151-1934-151-1943b. (Atifur)
Halt and resume schedule several times. Tried to reconfigure and reboot the DBBC. Received continuous alarm “Alarm large difference between formatter and maserdelay. Check for stability of new offset”. Delaylive show linear drifting. Missed scans 152-0022a to 152-0027a, 152-0054 to 152-0105, 152-0123 to 152-0138b (mas)
Disk VSN: NRAO-145/2000/| Data volume at beginning: 0.0 GB
18:30:00 UT experiment start okay (Atifur).